Critical evaluation of methods used to determine amplification efficiency refutes the exponential character of real-time PCR
© Rutledge and Stewart; licensee BioMed Central Ltd. 2008
Received: 12 June 2008
Accepted: 30 October 2008
Published: 30 October 2008
The challenge of determining amplification efficiency has long been a predominant aspect of implementing real-time qPCR, playing a critical role in the accuracy and reliability that can be achieved. Based upon analysis of amplification profile position, standard curves are currently the gold standard for amplification efficiency determination. However, in addition to being highly resource intensive, the efficacy of this approach is limited by the necessary assumption that all samples are amplified with the same efficiency as predicted by a standard curve. These limitations have driven efforts to develop methods for determining amplification efficiency by analyzing the fluorescence readings from individual amplification reactions. The most prominent approach is based on analysis of the "log-linear region", founded upon the presumption that amplification efficiency is constant within this region. Nevertheless, a recently developed sigmoidal model has provided new insights that challenge such historically held views, dictating that amplification efficiency is not only dynamic, but is linearly coupled to amplicon DNA quantity. Called "linear regression of efficiency" or LRE, this kinetic-based approach redefines amplification efficiency as the maximal efficiency (Emax) generated at the onset of thermocycling.
This study presents a critical evaluation of amplification efficiency determination, which reveals that potentially large underestimations occur when exponential mathematics is applied to the log-linear region. This discrepancy was found to stem from misinterpreting the origin of the log-linear region, which is derived not from an invariant amplification efficiency, but rather from an exponential loss in amplification rate. In contrast, LRE analysis generated Emax estimates that correlated closely to that derived from a standard curve, despite the fact that standard curve analysis is founded upon exponential mathematics. This paradoxical result implies that the quantitative efficacy of positional-based analysis relies not upon the exponential character of real-time PCR, but instead on the ability to precisely define the relative position of an amplification profile.
In addition to presenting insights into the sigmoidal character of the polymerase chain reaction, LRE analysis provides a viable alternative to standard curves for amplification efficiency determination, based on analysis of high-quality fluorescence readings within the central region of SYBR Green I generated amplification profiles.
All commercial real-time quantitative PCR platforms currently rely on defining the relative position of amplification profiles. As such, they are reliant on amplification of a serially diluted target to provide an estimate of amplification efficiency, which is essential to accurate and reliable quantification [1, 2]. However, a major caveat of this approach is that sample-specific inhibitors can compromise both the reliability and accuracy of an assay. This can be a major concern, particularly for samples derived from sources known to contain PCR inhibitors, in that any loss of amplification efficiency will generate unidentified and potentially large quantitative errors [3, 4].
Driven by the highly desirable ability to assess amplification kinetics within individual PCR reactions, a large number of studies have attempted to utilize the kinetics of amplicon DNA accumulation as the basis for amplification efficiency determination, through the application of exponential mathematics [5–15]. Derived from the intuitive notion that PCR amplification is inherently exponential in nature, this approach attempts to exploit the presence of a "log-linear region" within the lower region of real-time amplification profiles. Founded on the presumption that log-linearity reflects constant amplification efficiency, amplification efficiency is calculated from the slope of the log-linear region, similar to that conducted for standard curves.
An alternative approach to fluorescence analysis is based upon the presumption that PCR amplification is inherently sigmoidal in nature, allowing amplification efficiency to be estimated by fitting fluorescence readings to the classic Boltzmann sigmoid function using nonlinear regression analysis [16–26]. Unfortunately, effective implementation of this approach has been impeded by errors produced by distortions within the plateau phase [18, 23–25]. Recent development of a simplified approach to sigmoidal analysis based on recognition that amplification rate is linearly coupled to amplicon DNA quantity, circumvents such plateau phase anomalies. Called "linear regression of efficiency" or LRE analysis, amplification efficiency is determined by applying linear regression analysis to the fluorescence readings within the central region of an amplification profile .
Under the LRE model, amplification efficiency is maximal at the onset of thermocycling, with amplification rate progressively slowing as amplicon DNA accumulates, such that each cycle produces a unique amplification efficiency, with entry into the plateau phase occurring as amplification efficiency approaches zero. As such, amplification efficiency is defined as the maximal efficiency (Emax) produced in the absence of amplicon DNA. Although this clearly conflicts with the exponential model of real-time qPCR, which dictates that amplification efficiency is constant, it is unclear as to what extent these opposing interpretations impact the efficacy of amplification efficiency determination.
A central objective of this study was to critically evaluate exponential- and sigmoidal-based fluorescence analysis for determining amplification efficiency, with the expectation that standard curves would provide a gold standard from which to base the comparison. Notwithstanding the extensive resources required for their construction, the positional analysis upon which standard curves are based proved to be an effective platform from which to conduct the analysis.
Results and discussion
Ct-based standard curve analysis
where N 0 is the number of target molecules, N t is the number of amplicon molecules at threshold and E is the amplification efficiency. Target quantity can thus be calculated once values for N t and E have been obtained, which can be accomplished by constructing a standard curve in which Log(N0) is plotted against Ct, such that amplification efficiency is defined by the slope:
E slope = 10 slope - 1
where Eslopeis the slope-derived estimate of amplification efficiency . Figure 1B illustrates this approach, which produced an E slope of 92.0% with a linear correlation coefficient (r2) of 0.9997.
Such a high level of linearity is consistent with the contention that PCR amplification is exponential in nature, which in turn supports the presumption that amplification efficiency is invariant. Having said this, however, it is also evident from the sigmoidal shape of a typical SYBR Green I real-time profile that amplification efficiency is not completely invariant, as reflected by the eventual cessation of amplification that defines the end point of the amplification process, commonly referred to as the "plateau phase" (Figure 1A). This potential contradiction has led many prominent studies to surmise that an "exponential region" exists within the lower domain of amplification profiles [6, 9, 10, 13–15, 20, 28–30], with the inference that the upper boundary of this exponential region is defined as the point at which the amplification efficiency begins to decrease.
Not only does this contest the existence of an exponential region, it also challenges the perception that the efficacy of the threshold method is reliant on the exponential character of real-time amplification profiles. Partial resolution to this apparent paradox can be gained by considering PCR amplification to be an inherently sigmoidal process in which amplification efficiency is dynamic [18, 27]. Under such a scenario, it could be surmised that the efficacy of the threshold method is derived from the application of a purely positional-based analysis. A slope-based estimate of amplification efficiency (Eslope) would thus reflect the rate of amplification, not at Ct, but rather the maximal amplification efficiency (Emax) as defined under a sigmoidal paradigm (see below). Indeed, sigmoidal analysis provides insights that support such an interpretation.
Kinetic-based sigmoidal analysis
Recognition that amplification rate is linearly correlated with amplicon quantity led to the proposal that the dynamics of PCR amplification can be described by the linear equation:
E C = ΔE × F C +Emax
Although C1/2 may appear to be similar to Ct, a central attribute of C1/2 is that it is purely sigmoidal in origin. This then provides the opportunity to examine the efficacy of positional analysis by constructing a standard curve using the C1/2 values from each of the amplification profiles presented in Figure 1A. As presented in Figure 5B, this produces a slope-derived amplification efficiency estimate of 92.6%, a value that falls within the range of estimates of 92.0–92.6% produced by the Ct-based standard curve analyses (Figures 1B and 2B). This further supports the contention that the efficacy of the threshold method is dependent on accurately defining profile position, rather than on the exponential nature of real-time PCR.
Exponential analysis of the log-linear region
First introduced during the early history of quantitative PCR, a number of studies have attempted to determine amplification efficiency by exploiting the existence of a "log-linear region" within the lower region of an amplification profile [5–9]. Differing primarily in the number and position of the cycles included in the analysis, several implementations of this approach have been reported for real-time qPCR [9–15], which have subsequently been utilized for large-scale gene expression profiling in which standard curve construction is impractical [20, 31, 32]. Similar to the exponential mathematics of the threshold method, amplification efficiency is calculated from the slope of the log-linear region:
E loglin = 10slope - 1
This generated an average r2 of 0.9991, corroborating the presence of a substantive log-linear region within the lower region of all five amplification profiles. Nevertheless, the first indication of an anomaly is provided by the Eloglin values produced by the five profiles, which averaged 80.3% ± 2.5% (Figure 6). This is substantially lower than the Ct-Eslope of 92.0–92.6%, Emax of 95.5%, and C1/2-Eslope of 92.6% generated from the same five amplification profiles (Figures 1B and 2B, 3, and 5B, respectively).
An essential insight into the nature of the log-linear region is provided by LRE analysis, in that cycle efficiency can be calculated from the reaction fluorescence using equation 3. This predicts that a 15–30% loss in amplification efficiency occurs across the six cycles included in this analysis, despite the high level of linearity (Figure 6). In addition to challenging the contention that the log-linear region is representative of an exponential region, this predicted reduction in amplification efficiency would be expected to reduce the slope as compared with an invariant amplification efficiency, which in turn would reduce the resulting Eloglin values. Based on comparison with the "gold standard" Eslope values, this indeed appears to be the case. Ultimately, however, conflicting interpretations raise the question as to what is the true origin of the log-linear region.
Resolving the origin of the log-linear region
This confirms the presence of a log-linear region that generates an Eloglin estimate of 83.5%, similar to the 79.6% Eloglin estimate generated from analysis of the actual FC readings (Figure 6). LRE modeling predicts a large loss in amplification efficiency within this log-linear region in close agreement with that predicted from the actual FC dataset (Figure 6). Another key attribute of this sigmoid-derived log-linear region is illustrated in Figure 7B, which is that the loss in amplification efficiency is exponential in nature, suggesting that it is an exponential loss in amplification efficiency that underpins the linear character of the log-linear region.
Additional perspectives on the kinetics of real-time PCR are provided by Figure 7C, which demonstrates that loss in amplification efficiency can be estimated by expressing reaction fluorescence as a percentage of Fmax. For example, a reaction fluorescence that is 1% of Fmax corresponds to an approximate 1% loss in amplification efficiency. Importantly, this provides a simple method for illustrating at what point loss in amplification efficiency becomes nontrivial. In relation to SYBR Green I detection under the optical capabilities of the instrumentation used in this study, 1% of Fmax corresponds to about 100 fluorescence units, which is below the lowest fluorescence intensity that can be measured with acceptable accuracy. Accordingly, at 1,000 fluorescence units, which roughly corresponds to the middle of the log-linear region, the loss in amplification efficiency is estimated to be 10%. It would thus be difficult to obtain accurate fluorescence readings in which loss in amplification efficiency can be considered to be trivial, without a substantive increase in detection sensitivity. Of greater significance, however, is that a measurable loss in amplification efficiency occurs within 99% of a real-time amplification profile, irrespective of detection sensitivity, refuting the exponential character that has historically been ascribed to real-time PCR.
Comparison of automated data processing packages
Although detailed analysis of a single standard curve may be sufficient to demonstrate lack of exponential character, it only provides a limited perspective as to the potential impact on currently employed data processing packages. The evaluation was therefore expanded to include four identical standard curves in which the reaction mixes were supplemented with increasing quantities of SYBR Green I. Initially based on concerns that SYBR Green I quantity could impact the quantitative accuracy of LRE modeling , this approach also provides some perspective as to how each method responds to the inhibition of amplification, produced by increasing SYBR Green I quantity. It should also be noted that DyNAmo (formulated with an engineered T. brockianus DNA polymerase fused to a non-specific DNA-binding region) was used for this analysis, demonstrating that the trends described here are not unique to enzyme formulations containing T. aquaticus DNA polymerase.
Comparison of amplification efficiency determinations generated by six different methodologies, based on four replicate standard curves
Positional Analysis (Standard curve)
Fluorescence Analysis (Individual amplification profiles)
Serial Dilution (n = 6)
Linear Regression (E slope )
C t - E slope
LRE C 1/2 - E slope
LRE E max
(n = 6)
(n = 6)
"LinReg" (n = 6)
"Miner" (n = 6)
96.0% ± 1.7%
81.1% ± 1.6%
87.5 ± 3.5%
93.0 ± 1.8%
95.6% ± 1.3%
79.8% ± 1.6%
89.6 ± 3.0%
90.6 ± 2.1%
92.8% ± 0.9%
83.1% ± 1.4%
84.6 ± 4.2%
90.5 ± 2.8%
85.6% ± 0.7%
79.2% ± 2.1%
90.8 ± 3.8%
87.2 ± 1.7%
Similar to that observed previously, positional analysis produced a high level of correlation between Ct- and C1/2-based Eslope values. Both present a progressive loss in Emax as SYBR Green I quantity was increased, resulting in an 8% reduction at the highest quantity examined. Increasing SYBR Green I quantity had no apparent impact on the r2 of each standard curve, indicating that increasing reaction fluorescence did not improve the precision of the analysis.
As predicted by sigmoidal modeling (Figure 7) all three methods that use exponential analysis produced lower efficiency values, although the magnitude of this difference varied. Potentially more significant is that exponential analysis generated no discernible trend when SYBR Green I quantity was progressively increased, with the possible exception of Miner, which predicted a 6% loss at 2.0X SYBR Green I. It is important to also note that exponential analysis is highly influenced by the number and position of the cycles included in the analysis. This is particularly evident for LinReg in which cycle selection is based on searching for a maximal r2 using a variable cycle window size and position. This can produce large variances and frequently requires manual adjustment. However, regardless of implementation, the invalidity of applying exponential mathematics to the log-linear region brings into question both the reliability and accuracy generated by LinReg and Miner.
Consistent with that seen in Figure 3, automated LRE analysis produced Emax values similar to the Eslope values, except at the highest SYBR Green I quantity, which produced a ~4% lower efficiency estimate. Notwithstanding this discrepancy, this dataset does provide substantive supporting evidence that LRE analysis generates amplification efficiency estimates that correlate more closely to Eslope, and generate less variance than methods that rely on exponential analysis of the log-linear region.
Kinetic analysis of PCR amplification based on the LRE model reveals fundamental flaws in the current interpretation of amplification efficiency dynamics, demonstrating that SYBR Green I-derived amplification profiles lack the exponential character that has historically been ascribed to real-time PCR. Consequently, methods for amplification efficiency determination that rely on exponential analysis of the log-linear region generate systematic underestimations, differing only in the extent of the bias and variability in the resulting amplification efficiency estimates. Although this apparent lack of exponential character could also be expected to compromise the efficacy of the threshold method, empirical testing demonstrates that positional analysis does not rely on the exponential character of a real-time amplification profile. Furthermore, it was demonstrated that the slope-derived efficiency estimate produced by Ct-based standard curves reflect the amplification efficiency not at Ct, but rather the maximal amplification efficiency as defined under a sigmoidal paradigm.
This study further corroborates the efficacy of LRE analysis for amplification efficiency determination, as well as providing additional insights into the linear coupling between amplification efficiency and amplicon DNA quantity. LRE analysis thus not only provides a gateway to sigmoid-based quantification, but also provides a simple methodology for analyzing amplification kinetics within individual amplification reactions. Based on analysis of the high-quality fluorescence readings within the central region of an amplification profile, LRE analysis avoids errors associated with both low reaction fluorescence and distortions associated with the plateau phase.
PCR amplification was conducted as previously described  in which 3–4 replicate reactions were run for each quantity of lambda gDNA, and the FC datasets averaged to generate a single amplification profile for analysis. Briefly, replicate amplification sets consisting of 5.0 μl reactions containing lambda gDNA (New England BioLabs) at the specified quantity and 500 μM of the lambda primers K7B (CTGCTGGCCGGAACTAATGAATTTATTGGT) and K12 (ATGCCACGATGCCTCATCACTGTTG). The standard curve presented in Figure 1 employed QuantiTect (Qiagen) enzyme formulation, whereas DyNAmo (Finnzymes, distributed by New England BioLabs) was used for the standard curves containing increasing quantities of SYBR Green I (Table 1). SYBR Green I was diluted to the appropriate quantity using ddH20 before addition to the PCR master mix just prior to amplification reaction preparation and is expressed in units designated by the manufacturer (Invitrogen).
All amplifications were conducted with a Mx3000P spectrofluorometric thermal cycler (Stratagene) using a two temperature cycling regime initiated with a 15 min activation at 95°C, followed by 50 cycles of 120 s annealing and elongation at 65°C and a 10 s denaturation at 95°C. To increase optical precision, three fluorescent reads were taken at the end of the annealing and elongation step and the average used as an estimate of reaction fluorescence. Specificity of amplification was confirmed by melting curve analysis conducted at the end of each run.
An extensive description of the development and implementation of the LRE method is provided by Rutledge and Stewart . Automated LRE analysis was conducted using the prototypic Java program provided as supplementary materials in this earlier study using default values.
We are indebted to Frédéric Vigneault and Brian Boyle for critically reviewing the manuscript, and thank Pamela Cheers for editorial assistance. The research was supported in part by a grant to RGR from the Canadian Biotechnology Strategy.
- Pfaffl MW: A new mathematical model for relative quantification in real-time RT-PCR. Nucleic Acids Res 2001, 29(9):e45. 10.1093/nar/29.9.e45.PubMed CentralView ArticlePubMedGoogle Scholar
- Rutledge RG, Côté C: Mathematics of quantitative kinetic PCR and the application of standard curves. Nucleic Acids Res 2003, 31(16):e93. 10.1093/nar/gng093.PubMed CentralView ArticlePubMedGoogle Scholar
- Meijerink J, Mandigers C, Locht L, Tönnissen E, Goodsaid F, Raemaekers J: A novel method to compensate for different amplification efficiencies between patient DNA samples in quantitative real-time PCR. J Mol Diagn 2001, 3(2):55-61.PubMed CentralView ArticlePubMedGoogle Scholar
- Ståhlberg A, Åman P, Ridell B, Mostad P, Kubista M: Quantitative real-time PCR method for detection of B-lymphocyte monoclonality by comparison of kappa and lambda immunoglobulin light chain expression. Clin Chem 2003, 49(1):51-59. 10.1373/49.1.51.View ArticlePubMedGoogle Scholar
- Chelly J, Kaplan JC, Maire P, Gautron S, Kahn A: Transcription of the dystrophin gene in human muscle and non-muscle tissue. Nature 1988, 333(6176):858-860. 10.1038/333858a0.View ArticlePubMedGoogle Scholar
- Wiesner RJ, Rüegg JC, Morano I: Counting target molecules by exponential polymerase chain reaction: copy number of mitochondrial DNA in rat tissues. Biochem Biophys Res Commun 1992, 183(2):553-559. 10.1016/0006-291X(92)90517-O.View ArticlePubMedGoogle Scholar
- Wiesner RJ: Direct quantification of picomolar concentrations of mRNAs by mathematical analysis of a reverse transcription/exponential polymerase chain reaction assay. Nucleic Acids Res 1992, 20(21):5863-5864. 10.1093/nar/20.21.5863.PubMed CentralView ArticlePubMedGoogle Scholar
- Harting I, Wiesner RJ: Quantification of transcript-to-template ratios as a measure of gene expression using RT-PCR. BioTechniques 1997, 23(3):450-455.PubMedGoogle Scholar
- Gentle A, Anastasopoulos F, McBrien NA: High-resolution semi-quantitative real-time PCR without the use of a standard curve. Biotechniques 2001, 31(3):502.PubMedGoogle Scholar
- Liu W, Saint DA: A new quantitative method of real time reverse transcription polymerase chain reaction assay based on simulation of polymerase chain reaction kinetics. Anal Biochem 2002, 302(1):52-59. 10.1006/abio.2001.5530.View ArticlePubMedGoogle Scholar
- Ramakers C, Ruijter JM, Deprez RH, Moorman AF: Assumption-free analysis of quantitative real-time polymerase chain reaction (PCR) data. Neurosci Lett 2003, 339(1):62-66. 10.1016/S0304-3940(02)01423-4.View ArticlePubMedGoogle Scholar
- Peirson SN, Butler JN, Foster RG: Experimental validation of novel and conventional approaches to quantitative real-time PCR data analysis. Nucleic Acids Res 2003, 31(14):e73. 10.1093/nar/gng073.PubMed CentralView ArticlePubMedGoogle Scholar
- Bar T, Stahlberg A, Muszta A, Kubista M: Kinetic Outlier Detection (KOD) in real-time PCR. Nucleic Acids Res 2003, 31(17):e105. 10.1093/nar/gng106.PubMed CentralView ArticlePubMedGoogle Scholar
- Wilhelm J, Pingoud A, Hahn M: SoFAR: software for fully automatic evaluation of real-time PCR data. BioTechniques 2003, 34(2):324-332.PubMedGoogle Scholar
- Zhao S, Fernald RD: Comprehensive algorithm for quantitative real-time polymerase chain reaction. J Comput Biol 2005, 12(8):1047-1064. 10.1089/cmb.2005.12.1047.PubMed CentralView ArticlePubMedGoogle Scholar
- Liu W, Saint DA: Validation of a quantitative method for real time PCR kinetics. Biochem Biophys Res Commun 2002, 294(2):347-353. 10.1016/S0006-291X(02)00478-3.View ArticlePubMedGoogle Scholar
- Tichopad A, Dzidic A, Pfaffl MW: Improving quantitative real-time RT-PCR reproducibility by boosting primer-linked amplification efficiency. Biotechnol Lett 2002, 24: 2053-2056. 10.1023/A:1021319421153.View ArticleGoogle Scholar
- Rutledge RG: Sigmoidal curve-fitting redefines quantitative real-time PCR with the prospective of developing automated high-throughput applications. Nucleic Acids Res 2004, 32(22):e178. 10.1093/nar/gnh177.PubMed CentralView ArticlePubMedGoogle Scholar
- Goll R, Olsen T, Cui G, Florholmen J: Evaluation of absolute quantitation by nonlinear regression in probe-based real-time PCR. BMC Bioinformatics 2006, 7: 107. 10.1186/1471-2105-7-107.PubMed CentralView ArticlePubMedGoogle Scholar
- Karlen Y, McNair A, Perseguers S, Mazza C, Mermod N: Statistical significance of quantitative PCR. BMC Bioinformatics 2007, 8: 131. 10.1186/1471-2105-8-131.PubMed CentralView ArticlePubMedGoogle Scholar
- Roussel Y, Harris A, Lee MH, Wilks M: Novel methods of quantitative real-time PCR data analysis in a murine Helicobacter pylori vaccine model. Vaccine 2007, 25(15):2919-2929. 10.1016/j.vaccine.2006.07.013.View ArticlePubMedGoogle Scholar
- Qiu H, Durand K, Rabinovitch-Chable H, Rigaud M, Gazaille V, Clavere P, Sturtz FG: Gene expression of HIF-1alpha and XRCC4 measured in human samples by real-time RT-PCR using the sigmoidal curve-fitting method. Biotechniques 2007, 42(3):355-362. 10.2144/000112331.View ArticlePubMedGoogle Scholar
- Cikos S, Bukovska A, Koppel J: Relative quantification of mRNA: comparison of methods currently used for real-time PCR data analysis. BMC Mol Biol 2007, 8: 113. 10.1186/1471-2199-8-113.PubMed CentralView ArticlePubMedGoogle Scholar
- Durtschi JD, Stevenson J, Hymas W, Voelkerding KV: Evaluation of quantification methods for real-time PCR minor groove binding hybridization probe assays. Anal Biochem 2007, 361(1):55-64. 10.1016/j.ab.2006.11.023.View ArticlePubMedGoogle Scholar
- Swillens S, Dessars B, Housni HE: Revisiting the sigmoidal curve fitting applied to quantitative real-time PCR data. Anal Biochem 2008, 373(2):370-376. 10.1016/j.ab.2007.10.019.View ArticlePubMedGoogle Scholar
- Feng J, Zeng R, Chen J: Accurate and efficient data processing for quantitative real-time PCR using a tripartite plant virus as a model. Biotechniques 2008, 44(7):901-912. 10.2144/000112750.View ArticlePubMedGoogle Scholar
- Rutledge RG, Stewart D: A kinetic-based sigmoidal model for the polymerase chain reaction and its application to high-capacity absolute quantitative real-time PCR. BMC Biotechnol 2008, 8: 47. 10.1186/1472-6750-8-47.PubMed CentralView ArticlePubMedGoogle Scholar
- Tichopad A, Dilger M, Schwarz G, Pfaffl MW: Standardized determination of real-time PCR efficiency from a single reaction set-up. Nucleic Acids Res 2003, 31(20):e122. 10.1093/nar/gng122.PubMed CentralView ArticlePubMedGoogle Scholar
- Schefe JH, Lehmann KE, Buschmann IR, Unger T, Funke-Kaiser H: Quantitative real-time RT-PCR data analysis: current concepts and the novel "gene expression's C T difference" formula. J Mol Med 2006, 84(11):901-910. 10.1007/s00109-006-0097-6 10.1007/s00109-006-0097-6View ArticlePubMedGoogle Scholar
- Chervoneva I, Li Y, Iglewicz B, Waldman S, Hyslop T: Relative quantification based on logistic models for individual polymerase chain reactions. Stat Med 2007, 26(30):5596-5611. 10.1002/sim.3127.View ArticlePubMedGoogle Scholar
- Czechowski T, Bari RP, Stitt M, Scheible WR, Udvardi MK: Real-time RT-PCR profiling of over 1400 Arabidopsis transcription factors: unprecedented sensitivity reveals novel root- and shoot-specific genes. Plant J 2004, 38(2):366-379. 10.1111/j.1365-313X.2004.02051.x.View ArticlePubMedGoogle Scholar
- Caldana C, Scheible WR, Mueller-Roeber B, Ruzicic S: A quantitative RT-PCR platform for high-throughput expression profiling of 2500 rice transcription factors. Plant Methods 2007, 3: 7. 10.1186/1746-4811-3-7.PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.